Express - Calibration
- Ensure load cell ratings (Rated Capacity and Sensitivity) have been found and entered in the Device tab.
- Ensure the Libra Express device and app are paired and communicating.
- Ensure the equipment is empty and on level ground.
- Select the Device tab and tap Zero Scale.
- Select the Display tab and then tap Tare.
- Load the equipment with a known weight and note the reading from the large display in the app. Larger weights provide higher precision.
- Select the Device tab and enter the known weight as the Certified Reading.
- Enter the noted amount as the Libra Reading.
- Tap Calculate to compute a new Calibration Factor.
- Tap Save Settings.
No results found